What P is the time taken to complete one waveform. What F is the number of waveforms produced in one second. What P is a voltage measurement from one extreme of the amplitude to another. What T is measured by the x-axis of the scope display. What T is required to capture free running signals. What W is selected using the function generator. What H is the unit of frequency. What C should take place before you connect a signal from the function generator. What D is a type of scope with an "AUTOSET" facility. What M is one thousandth of a second. What M is one millionth of a second. What S is common calibration signal. What C allows a scope to display not just one but two signals. What B is connected to the channel so that a 4mm lead can be used. What D is the oscilloscope screen divided into. What P supplies Vrms voltage. What A comes out of the mains socket. What F is the UK mains frequency. What I controls the brightness of the oscilloscope. What F control may be adjusted to produce a clear signal.
If you are seeing this message then you do not have Adobe Flash Player installed. To see thin interactive game you will need to
the latest version of Adobe Flash Player.
Click here for more free